|Place of Origin:||China|
|Minimum Order Quantity:||1 set|
|Delivery Time:||20 days|
|Payment Terms:||L/C, T/T, Western Union|
|Supply Ability:||20 sets per month|
|Test Standard:||ASTM D92,GB/T3536,ISO2592||Measuring Range:||Ambient~400℃|
|Purpose:||Measure Flash Point Of Petroleum Products||Powr Consumption:||400W|
According to test method of GB/T3536,this open cup flash point analyzer adopts advanced Microprocessor and PID temperature controlling technology, the temperature rising speed is fast, temperature controlling is more precise, and test result is more accurate.
LCD big screen replaces old design of RS323 port, the operation is easier. This machine is highly automatic, which can automatically calibrate atmosphere, lift, rise temperature, switch, ignite, inspect final point, print test result, air cool etc. All test procedures are automatic, and it can stop when temperature exceeds limit.
1.The open cup flash point tester is fully automatically controlled by micro computer, all procedures and result printing are automatic.
2.Big screen interface displays test temperature, oil serial number, test time etc. All process is clear.
3.The micro-printer can save cost.
4.Tempeature is raised by SCM program, and it makes sure the result correct.
5.The flash point tester is automatically cooled by air when the test is finished,
6.The flame size can be adjusted.
7.Automatic lifting system is easy for operation
8.It’s made by cold-rolled sheet, surface treated by Electrostatic spraying. It’s corrosion resisting and easy to clean.
9.The open cup flash point analyzer can be operated manually or automatically
10.Correction function: current date, time, temperature and atmosphere
11.It has protection and self-diagnosis procedures
12.Both main controlling and measuring parts are imported, which makes sure the reliability and stability.
13.The flash point analyzer is compact, small and beautiful
14.High degree of automation: set flash point as per operation method on LCD display screen, press “Start” key, the whole test will be proceeded automatically.
|1||Applicable standards||GB/T3536,ISO2592,ASTM D92|
① controlling temperature and expected set flash point
② Fault prompt
|6||Refrigeration method||forced air cooling|
|7||Environmental condition||Temperature:10～55℃;Humidity: 30～80%|
|8||Error||Flash point 6 ℃;Fire point 6℃|
|9||Repeatability||Flash point 12℃;Fire point 10℃|