|Place of Origin:||CHINA|
|Minimum Order Quantity:||1 SET|
|Packaging Details:||Plywood case|
|Delivery Time:||90 days|
|Payment Terms:||Western Union, T/T, L/C|
|Supply Ability:||10 sets per month|
|Test Standard:||EN 6942||Application:||Heat Radiant Of Protective Gloves And Protective Clothing|
|Required Gas:||Compressed Air||Warranty:||1 Year|
|Size:||1200*500*1300mm3||Power Supply:||AC220V，15000W ,Air Switch C80|
|Heating Flux:||10～80 Kw /m2|
It is suitable to determine heat radiation resistance or thermal conductivity of protective gloves and protective clothing.
1.Radiation source is made up of 6 groups heating bar,length of heat bar is 356mm±2mm,and the length of heating part is 178±2mm,diameter of heating bar is 7.9±0.1mm.
2.Radiation source uses PID temperature controller,controlling accuracy is ±1℃.
3.Heating capability of radiation source: Max.1200℃±5%,adopt PID adjustable controller.
4.Radiation heat flux:10～80 kw /m2
5.Sampling frequency if data acquisition card can reach 20Hz.
6.It has heat radiation flux sensor with quartz window and water circulating device,maximum measuring range is 100kw/m2.
7.It’s supplied with copper calorimeter,which can measure the temperature change of sample’s back.
8.Copper calorimeter is made of high conductivity oxygen-free pure copper.
9.Supplied with A and B type sample holders.
10.Sample moving device can be moved on platform freely.
11.Test parameter can be set by software,and it starts test automatically.It can test temperature rising 12s and 24s,heat flux density Q and heat transfer coefficient HF.
|1||Heat auxiliary facilities||Stainless steel and 2mm SS sheet machines,seamless welding|
|2||Sample holder||Stainless steel machines as per standard|
|3||Other auxiliary facilities||Aluminum or iron baking finish|
|4||Auxiliary fixtures||Aluminum processing or stainless steeel|